Rs.900 + Gst @12%
|Parameter Measures||SpO2 + Pulse Rate|
|Model Name/Number||FG 2.0|
|Display Mode||3 Display Modes|
|Oximeter Accuracy||Accuracy +/- 1BPM or +/-1 % (the larger one)|
|Weight||Weight: 1.8 ounce (50 gram)|
Features:1. A Battery life indicator2. In the absence of signals, the product would power-off automatically after 8 seconds.3. Small, light in weight, and convenient to carry.
Keys function description: In standby mode, start the keyinstrument into the working state, Touch this button underthe working state can change the display mode.Note : The machine profile picture is only for reference use,specificity in kind prevails. Parameter Setup:Press start button (> 3s), into parameter setup, As menu 1:1. In menu 1. Short press (< 1s) to switch the UI interface direction and Long press to enter thesetting mode of blood oxygen (> 3s)2. Put the finger into the test position of the oximeter, and the oximeter will automatically testand display the resultsAttentions• Using devices outdoors or under strong light; please adjust the screen brightness to ahigher level for proper observation.. It''s better for users to choose a lower brightness to conserve battery power.2. In menu 2. When the " *" signal is shown on the " Sounds Setup", Press the button (> 3s)and enter into the menu 1. Press the button (< 1s) to select an item, and thenPress the button (> 3s) to setup data. Choose " +" or "-" to increase or decrease values.
Operation Instructions1. Install a pair of AAA batteries into the battery compartment.2. Plug a finger into a rubber hole of the Oximeter.3. Press button on the front panel to start.4. Do not shake your finger or move your body when the Oximeter is working.5. Press the button if you want to change display direction.6. Read the relevant value from the display screen7. The device will automatically power off in 8 seconds if there is no finger signal.When plugging your finger into the Oximeter, your nail surface must be upward.Declaration: Please use the alcohol to clean the rubber before each test andclean the tested finger with alcohol before and after the test. (The rubber insideof the Oximeter is a rubber which has no toxin, causes no harm, and brings noside effect such as allergy to the skin).
Hang Rope Installation1. Put the rope thin end through the hole.2. Put the rope coarser end through its already wearing thin end part and tighten,
Product AccessoriesOne user manual One lanyard QC cardDetailed Descriptions of Product Functions:1. Display Type OLED display2. SpO2 Measurement range 70 %-99 ¬curacy +/-2 % on the stage of 80 % 99 %;+/- 3 % on the stage of 70 % -79 %;Below 70 % no requirementResolution: +/- 1 %3. PR Measurement range 30BPM-240BPMAccuracy +/- 1BPM or +/-1 % (the larger one)4. Power two AAA 1.5V alkaline batteries5. Power consumption below 30mA Diagram of Operation PrincipleAutomatic power-off: The product power off by itself whenno finger is in the product for 28 seconds
7. Operation Environment.Temperature 5 ° C 40 ° CHumidity 15 % 80 %Atmospheric pressure: 70kPa-106kPa
Transport, storage environment:Temperature: -10 ° C ~ 55 ° CHumidity: 595 %Atmospheric pressure: 50kPa-106kPa, non-corrosivegas and well-ventilated environment.